This manual provides documentation for the following instruments: EA (3 Hz – GHz). EA (3 Hz – GHz). EA (3 Hz – GHz). EA (3. This manual provides documentation for the following instruments: PSA Series. EA (3 Hz – GHz). EA (3 Hz – GHz). EA (3 Hz – GHz). This manual provides documentation for the following instruments: EA (3 Hz – GHz). EA (3 Hz – GHz). EA (3 Hz – GHz). EA (3.
|Published (Last):||18 May 2017|
|PDF File Size:||20.99 Mb|
|ePub File Size:||1.84 Mb|
|Price:||Free* [*Free Regsitration Required]|
Viewing A Signal Viewing a Signal 1. Don’t show me this message again.
AGILENT TECHNOLOGIES EA GETTING STARTED MANUAL Pdf Download.
You can order the service documentation for Agilent spectrum analyzers through your Agilent Sales and Service office. Press If the previously performed alignments did not mabual the problem, press Alignments Restore Align Defaults Is the analyzer displaying an error message? Page 23 Item Name Access features used with all analyzer modes and affect the state of the entire spectrum System Keys analyzer.
Increments or decrements the e4404a value. Save the file 1. Available only at time of purchase Chapter 5 Description Provides the factory calibration test data and the standard commercial calibration certificate at initial shipment.
Page 53 NOTE You should notice that the trace you saved, in this example trace 3, is loaded to trace 2. Making a Basic Measurement This chapter is divided into the following sections: Agilent handheld spectrum analyzer user’s guide pages. Sales and Service Office and are listed below. Each softkey More has several characters on it.
Item Name Access features used with all analyzer modes and affect the state of the entire spectrum System Keys analyzer. Restores the analyzer to its factory-defined state.
In this example you are loading a trace file; Don’t have majual account? File Menu Functions File Menu Functions This chapter describes how to use the functions located under the front-panel Data storage manuql retrieval are handled similarly to that of personal computers PCs.
Page 44 Viewing Catalogs and Saving Files The analyzer stores and retrieves data similarly to the way that a personal computer PC does: EaEaEaEa. Reset the analyzer by pressing 2. To be implemented in future firmware upgrade. AC Power ,anual Chapter Please be as specific as possible about the problem.
Page 34 Making a Basic Measurement This chapter is divided into the following sections: In Case of Difficulty This chapter includes information on how to check for a problem with your Agilent Technologies spectrum analyzer, and how to return it for service. Check to see if the green LED above the power switch is on. Installation and Setup Turning on the Analyzer for the First Time If using LAN, set the IP address of the analyzer to an appropriate number for your network one that the network recognizes, but that is not yet in use: Select the type of file you want to save.
Initial Inspection Initial Inspection Inspect the shipping container and the cushioning material for signs of stress. Page 25 Item Description Data invalid indicator Status message area Key menu title Softkey menu Frequency span or stop frequency Sweep time Video bandwidth Frequency offset Display status line Resolution bandwidth Center frequency or start frequency Internal preamp Auto alignment routine status Trigger or Trace Average Display line annotation Firmware Revision To view the firmware revision of your analyzer, press call Agilent Technologies regarding your analyzer, it is helpful to have this revision and the analyzer serial number available.
Agilent Technologies e4440w not be liable for errors contained herein or for incidental or consequential damages in connection with the furnishing, performance, or use of this material.
These features are only available when special options are installed.
Used Agilent EA
Turning on the Analyzer for the First Time Plug in the power cord. Making a Basic Measurement This chapter provides information on basic analyzer operation. This function can be auto-coupled. Page 18 When you have completed the printer setup for the analyzer, press the front panel key.
E4440A PSA Spectrum Analyzer, 3 Hz to 26.5 GHz
Page 80 Index personality options not in instrument power amplifiers preamplifiers preset factory types user, creating printer setup probes AC and DC product information rear panel symbols rear panel features reference level, setting reference, turning on 10 MHz rename files RF limiters By pressing ‘print’ button you will print only current page. The entry fields show the parameters for the files viewed. Firmware Revision Firmware Revision To view the firmware revision of your analyzer, press call Agilent Technologies regarding your analyzer, it is helpful to have this revision and the analyzer serial number available.
Options And Accessories Options and Accessories This chapter lists options and accessories available for your analyzer. They are only designated by the Name: If the printer is ready and the printer setup was successful, a printout of the analyzer display will be printed.
To see that trace 2 is in view mode, blank traces 1 and 3 by pressing.